Drill Unit & Handpieces
|Electric Drill System (Universal)*||260-028-150|
|Motor for Drill Unit||260-028-425|
|20:1 High Torque Handpiece||260-028-201|
|20:1 Low Torque Handpiece||260-028-202|
*Only the 20:1 High Torque Handpiece (260-028-201) is necessary when using this drill unit.
- The 400:1 handpiece has sufficient torque to cut through bone at 50 RPM.
- The 20:1 handpiece is designed to be used with the pilot drill at ~1,100 RPM with external irrigation.
- Irrigation is not necessary at speeds of 50 RPM or lower.
- Failure to observe speed limitation may result in burned or overheated bone which could result in substantial bone necrosis.
- The 20,000 RPM Air Motor attaches to conventional four hole dental tubing and is an alternative to the electric drill unit.
- Air pressure of at least 80 PSI is recommended to drive the air motor with sufficient torque in dense bone.
- Proper maintenance of handpieces is crucial for long term success of handpieces. See reference.
- Do not exceed 135°C (275°F) when sterilizing.
- In order to prevent discoloration and/or damage to the plating of the handpieces from chemicals that are not sufficiently cleaned from other instruments, do not autoclave the handpieces with other instruments.
- Always check the handpiece for any abnormal vibration, heating, noise, or sluggish operation. If any abnormality is noticed, cease the use of the handpiece.